[ List Subjects ][ Main Page ]
Subject: Re: calibrated meter
[ View Followups ][ Post Followup ]
Date: 01/22/02 at 4:18 PM
Posted by: Russ Kincaid
In Reply to: calibrated meter posted by Jim on 01/22/02 at 2:53 PM:
How are you measuring power? If you are using DC meters, the error is no doubt due to the difference between average, which the meter reads, and RMS, which is the true power. You would do better to not convert to DC, but use true rms meters. Alternativly, you can calculate the difference between average and RMS and re-calibrate your meters.
Post a Follow-up: