[ List Subjects ][ Post Message ]
Subject: Re: Silver E-beam evaporation
[ View Followups ][ Post Followup ]
Date: 10/13/03 at 3:52 PM
Posted by: jim
In Reply to: Silver E-beam evaporation posted by Gord Wang on 10/02/03 at 12:37 PM:
this is not normal. you should use a moly or Aluminum oxide liner with silver not graphite. The moly liner is easier to use as the alumina liner is a insulator and requires a more difficult start up. You might also want to make sure your vacuum is good <5x10-4 Torr and that you do not have any leaks in the system.
Post a Follow-up: