In Reply to: Re: FS: Tektronix 465 & HP O-Scopes posted by Alejandro Flores on 03/27/00 at 1:06 PM:
: Could you please check this one as well as the price.
It checks out fine - as well as I can check it. That means that I have checked both channels
by connecting the probe to my AC line. Got a nice clear sine wave. I don't have the
capability to check all the other features. However, the HP came from the same co.
as the Teks. I have sold 6 of those, 2 to the same guy and my responses from them
have all been very favorable. I don't expect any problems with this one either.
However, if it doesn't work properly I will be happy to take it back and refund your purchase
I would like to get $250 for it.
: 5) HP1746A 100Mhz, 2 chnl. w/ time interval multimeter built-in on
: top. This appears to be like the DM44 on a TEK scope. New 10x probe.
: No cover or manual. Calibration date 5/8/98. I haven't had time to
: make sure this one is OK. So if you are interested, I will check it